The IFC signal converter is designed to measure the flow velocity, IFC signal converter can be combined with any KROHNE flow. The Krohne IFC signal converter will need to be setup to output pulses from the flow meter. This guide will outline the procedure on the IFC signal. Ifc • Read online or download PDF • KROHNE IFC Converter Quickstart EN User Manual.
|Published (Last):||15 June 2013|
|PDF File Size:||4.89 Mb|
|ePub File Size:||11.71 Mb|
|Price:||Free* [*Free Regsitration Required]|
Send us your email, and we’ll send you the further instructions. No measurement can be performed, the flow is set to zero. Multi-drop mode Yes, current output If “standard” is set, set field frequency in Fct. F backup 2 settings. Load on current output A too high, effective. Ambient temperature Depending on the version and combination of outputs.
Complain wrong Brand wrong Model non readable. F backup 2 settings F mqnual A The current at the current input is less than 0.
F factory settings Error during the CRC check of the factory settings. F flow exceeding limit Measuring range exceeded, filter setting limits measured values. Measuring pipe not filled, function. You have one of those, right? No message if empty pipe.
ADC over-ranged by DC offsets.
IFC – KROHNE Messtechnik – PDF Catalogs | Technical Documentation | Brochure
Open the catalog to page 5. Group message, when errors as described below or other application errors occur. Error during the CRC check of the active. F backup 1 settings. Open the catalog to page 4.
The difference lies in.
KROHNE IFC Converter Quickstart EN User Manual | 32 pages
The difference lies in whether the measured value is also set to zero upon the detection of an empty pipe. As with any electromagnetic flow meteryou need to install it carefully. The two empty pipe messages cannot appear at the same time.
Current not correct, mA output cable has open circuit or load too high. Measuring system Measuring principle Application range Continuous measurement of current volume flow, flow velocity, conductivity, mass flow at constant densitycoil temperature of the manusl sensor Design Modular construction The F field frequency too high Field frequency is not reaching steady state, a measured flow value is still being supplied but may have errors.
Operating elements 4 push buttons for operator control of the signal converter without opening the housing. Display if user interface Graphic display LC display, backlit white. Open the catalog to page 6.
Unsere Lieferzeiten während der Feiertage
F krohnw range A The pulse rate or the corresponding measured value is limited by a filter setting. Open the catalog to page Group message, when errors as described. Upload backup 1 or backup 2 settings, check and adjust if necessary.
F kroohne settings Error during the CRC check of the active settings. Now create your password. Application-dependent fault, but device is. Application test or operator action necessary. Application fault, device OK, but measured values affected.
Error during the CRC check of the backup 1. Forgot user name or password? The current at the current input is less than. A signal cable is used to connect the measuring sensor and the converter for the purposes of power supply and signal processing.
Current not correct, mA output cable has. Or the demanded pulse rate is too high.