Technical Spares. User Manual, GSI Product Code: Other Product Code: Manufacturer: Grason Stadler. Instruction Manual, GSI Direct calibration for TDH headset and bone vibrator. THE COMPLETE CLINICAL AUDIOMETER. AND MORE. The GSI 61 comes with everything you’ll need to. 2 GSI 61 NEW STANDARD IN CLINICAL AUDIOMETRY The GSI 61 is the versatile, making hearing aid evaluation easier than ever Direct calibration for TDH.
|Published (Last):||28 December 2015|
|PDF File Size:||5.98 Mb|
|ePub File Size:||19.94 Mb|
|Price:||Free* [*Free Regsitration Required]|
When the LCD is in the lowered position, easy access to the rear connector panel is provided. We are committed to giving you the best home security available today More information.
Present word discrimination material at 50 dB HL through Channel 1 to the aided test ear. Record the level at which the patient judges both of the signals to be of equal loudness. Page 70 Stop Bits: Since 19, we have provided knowledge, tools. B These records are sent by the GSI 61 to the remote computer to transfer parameter or status information. The Allegro offers quick and easy testing with a familiar four button navigation.
The operator presses the Correct or Incorrect pushbutton after each presentation, depending on the response. Present the simultaneous tones. B Chapter 6 – Remote Options Rev. When the timer is first selected, it initializes to 0: Exploring the value of EPIQ premium ultrasound Reducing cost of care through more definitive diagnoses Philips EPIQ offers a new level of premium performance ultrasound, and with it the opportunity to. Equipment that simply works, every time.
GSI 61 – Grason-Stadler – PDF Catalogs | Technical Documentation
With the status display, the GSI 61 is brought to the initialization state. The channel turns off immediately when the bar is released. Determine the threshold level for each ear at all frequencies being tested. Pressing Clear after Stop resets the timer to 0: The dashed grid lines are displayed at the intermediate frequencies of 9 kHz, Generally, the serial port parameters baud rate, stop bits, parity, handshaking, etc.
If not, this error message will be flashed: Masking noise may audiometef routed to the contralateral ear through Channel 2.
Prepared for the shopfloor our. While Talk Forward mqnual on, the only control which can be activated is the Print, if available.
This occurs while processing a completed input command, while printing, while the GSI 61 is transmitting a record and whenever the remote device is not enabled through the Remote button. Turn on the GSI 61 and allow it to warm up for 10 minutes. This display indicates the current instrument state and VU meters.
It lets you perform air and bone conduction testing with pure tone and live voice speech testing while monitoring the test results on the status or audiogram displays. Select the gsl output transducer routing to the desired ear.
GSI 61 Audiometer CLINICAL TWO-CHANNEL AUDIOMETER – PDF
Accessory equipment connected to the analog and digital interfaces must be certified to the respective IEC standards IEC for data processing or IEC for medical equipment.
Chapter 2 – Installation Insert Phone Outputs Monaural phone jacks for left and right insert phones or patch cords. The SISI percent score is displayed on the status screen.
Bsi diagnostic tests are easily selected through use of the ergonomically designed front key panel. The minimum contralateral interference is determined in the same audiomdter previously described using spondees.
GSi 61 User Manual
Proceed as with pure tone threshold testing, recording the thresholds onto the audiogram. When ALTernate is selected and the tone bar is released, the complete presentation of both channel signals is completed. This can be rectified by simply powering down for approximately thirty seconds and then restarting operation. Try the same combination on the alternate channel. Contrast This rotary control is used to set the brightness of the LCD.
Hear For Life view seller’s other items: The record specifies the control operation using a function code which defines the group of controls and a sub-function code which defines the specific parameter to select of the function to perform.