HANDLING THIS SERVICE MANUAL About This Service Manual n. Scope This Service Manual applies to the Fuji Medical Dry Laser Imager DRYPIX Service Technicians PACS / DR / CR / Digital Forum Drypix Service Manual CARLOS RUEDA Innovacion Tecnologica Hello, I am trying. View and Download FujiFilm DRYPIX service manual online. DRYPIX Printer pdf manual download.
|Published (Last):||23 February 2013|
|PDF File Size:||6.93 Mb|
|ePub File Size:||3.65 Mb|
|Price:||Free* [*Free Regsitration Required]|
Page When reinstalling the adiabatic cover, do not sandwich the thermistor cable. Add to My Bench. Page 94 26 x 36 cm size 20 x 25 cm size Conveyor Unit Guide Remove the conveyance driving roller lower.
Fujifilm – DryPix Community, Manuals and Specifications | MedWrench
Page MC 7. After using the manual, keep it nearby for future reference. Page 87 Remove the sub-scanning unit fixing bracket, and reinstall it to its original position.
Got it, continue to print. Page MCT board. Lower Conveyor Unit Remove the lower conveyor unit. If so, circle the out of control data points, correct the cause of the problem refer to the Performance and Corrective Action section of this document and repeat the test.
DryPix Operation Manual. Film Positioning Motor upper md5 Remove the upper conveyor unit. Documents Flashcards Grammar checker. Artifacts Banding, Scratch, Fog, etc. Fujifilm DryPix Specification 4 pages. Remove 400 gear 1 and gear 2. Comments to this Manuals Your Name. Artifacts Output test pattern films to make sure that there are no clinically significant artifacts, such as banding, scratch, fog etc.
Heat Development Unit Remove the film release unit.
FUJIFILM DRYPIX SERVICE MANUAL Pdf Download.
Geometry Recording Length and Width 4. Due to advances in technology, the regulation and testing of My Bench Order History Sign out.
The x-ray detector system is fundamental to the performance of Reboot the equipment, and start the M-Utility. Not only does each mammographic x-ray NOTE The following path accesses the step test pattern: Remove the special screws. The screws painted in yellow must be adjusted after the installation of parts. Secure the cable with the clamp.
Removal Drive Cam Remove the gear 1 and gear 2.
Page 55 MC Remove the sub-scanning unit fixing bracket. The DryPix has excellent throughput between and films per hour making it ideal for modalities such as high speed CT.
To return the film size from 20 x 25 cm to serfice sizes, refer to the following figure and change the attaching position of the film release plate. Disconnect the network cable of the PC for servicing from the network to which the equipment is connected.
However, perform the following when securing the sub-scanning unit into the equipment. Page 24 Reinstalling Procedure Reinstall in the reverse order of removal. When reinstalling the adiabatic cover, do not servive the thermistor cable.
EPS If the measured values are not the specified values, adjust the film conveyance amount. Drpix the belt cover. Squeezing Roller When removing the squeezing roller, be careful not to damage the bearing of the tray, Remove the squeezing roller.
FujiFilm DRYPIX 7000 Service Manual
EPS Secure the gear fixing cover with the screws. Tray Loading Unit Steps that differ between the three units are indicated separately.
Press the LED cover of the density measurement light-emitting board against the position shown in the figure below, and secure. Page 90 Remove the sub-scanning unit fixing bracket. Page Press the LED cover of the density mnaual light-emitting board against the position shown in the figure below, and secure. Removing Procedure Remove the following covers. Page MC Remove the cooling fan assembly. MC Remove the suction cup arm. Prepare the PC for servicing.
Steps that differ between the three units are indicated separately.